Testing Extruded Polystyrene Thermal Conductivity at Various Thicknesses with the Thermtest Heat Flow Meter
Figure 1. Acrylic plaster being applied on EPS insulation boards, on an apartment building façade.1
Extruded Polystyrene is a highly insulative construction material, generally installed on the exterior of above-grade studded walls, or the interior of foundation walls. Thus, knowing extruded polystyrene thermal conductivity values is important when determining the insulative potential of a building. Its purpose is to serve as a defense mechanism for heat loss in buildings, aiming at reducing the cost of operation. Extruded polystyrene is often confused with expanded polystyrene. Although similar in some aspects, such as their composition (polymerized polystyrol), these two insulation materials are vastly different. Extruded polystyrene is created by a process referred to as extrusion, hence the name. During this process, the polystyrol material is extruded through a die, causing the material to expand into a uniform closed-cell insulation board (Figure 2). Expanded polystyrene on the other hand is created by placing small foam beads into a mold and applying steam to expand the beads to form an insulation board (Figure 2). The manufacturing process of expanded polystyrene creates voids between the foam beads, creating pathways for potential moisture to enter.
Figure 2. Microscopic differences between the composition of extruded (left) and expanded (right) polystyrene foam insulations.2
Extruded Polystyrene Thermal Conductivity
The Thermtest Heat Flow Meter (HFM) is a steady-state heat transfer system measuring the thermal conductivity and thermal resistivity of flat slab insulating materials such as foams, solids and textiles (Figure 3). The HFM measures thermal conductivity over a range of 0.005 to 0.5 W/m·K and over a temperature range of -20°C to 70°C, in accordance with ASTM C518-15 – Standard Test Method for Steady-State Thermal Transfer Properties by Means of the Heat Flow Meter Apparatus. Users can expect a high degree of accuracy (3%) and precision (0.5%) with this ASTM traceable measurement technique.
Figure 3. Thermtest Heat Flow Meter (left) and samples of various thicknesses to test extruded polystyrene thermal conductivity (right).
According to the ASTM C518-15, to ensure the proper performance of the HFM, the apparatus must be calibrated with materials having similar thermal conductivities and thicknesses as the materials to be evaluated. If the calibration standard is tested at one thickness, the heat flow meter apparatus can be calibrated for that thickness. However, if tests are to be performed at varying thicknesses other than the calibrated thickness, a thorough study of the error of the HFM at other thicknesses must be conducted. For this experiment, Thermtest researchers set out to test the accuracy boundaries of the HFM, by testing multiple sample thicknesses based off of only one calibration thickness.
To begin, the calibration sample (NIST SRM 1450d – 1″) was placed between two parallel plates within the HFM (Figure 4). A specified temperature gradient (10 – 30°C) across the plates was established to simulate heat loss from the internal environment of a building, to the colder external environment. The upper plate was then set to compress onto the specimen to an automatic sample thickness. The HFM automatically determines sample thickness by employing four digital encoders, located in each corner of the upper plate. Each digital encoder measures the thickness at their post, and then the average is calculated. The upper plate then automatically adjusts to the average height. If the testing sample is highly compressible and an approximate compression force is known, a manual thickness setting may be a more suitable option in receiving accurate and precise thermal conductivity results.
Figure 4. View from inside the HFM door. Parallel plates (red and blue) establish a one-dimensional heat flux through the test specimen, simulating heat loss from a building’s interior to the external environment.
At constant, but distinct temperatures, the parallel plates established a steady state, one-dimensional heat flux through the test specimen and the thermocouples embedded in each plate, measured the temperature on either side of the calibration specimen. The heat flux transducers, in contact with the upper and lower plates, collected data on the resulting heat flux of the test specimen (Figure 4). By appropriate calibration of the heat flux transducer(s) with standards, and by measuring the plate temperatures and plate separation, Fourier’s law of heat conduction is used to calculate thermal conductivity (λ):
After the calibration was performed as mentioned above, each thickness of the extruded polystyrene sample was tested as per the steps outlined above.
The objective of this experiment was to determine the accuracy of the Thermtest HFM to test the thermal conductivity of varying thicknesses of extruded polystyrene at 20°C, with a single calibration sample. Thermal conductivity readings received from tests performed on thicknesses varying between 10.1 mm and 40.4 mm were within reason of the thermal conductivity reading of the 25.2 mm control test (less than 3%) (Figure 5). The results achieved correlate with results received from an experiment performed by Al-Ajlan in 2006, as well as the manufacturer provided data.
Figure 5. Thermal conductivity and thermal resistance values of various thicknesses of extruded polystyrene, calibrated against one layer of NIST SRM 1450d and performed at a mean temperature of 20˚C, using the Thermtest HFM.
Al-Ajlan (2006), reports the manufacturer provided thermal conductivity of expanded polystyrene to be 0.034 W/mK. This thermal conductivity is slightly higher than the manufacturer provided thermal conductivity of extruded polystyrene (0.032). Although extruded polystyrene has a lower thermal conductivity, meaning it is more likely to protect your internal environment from unwanted temperature changes, it has a significantly higher cost than the expanded polystyrene. Careful consideration must be taken into account when selecting the proper foam insulation board for your construction needs.
The Thermtest HFM is a fast, reliable and flexible method for testing thermal conductivities of solids, foams and textiles. Although not intended for use of testing various sample thicknesses with a single calibration thickness, this research proves the ability of the HFM to test the thermal conductivities of samples with slight variations in thicknesses, compared to the calibration sample.
Learn More About Thermtest The Heat Flow Meter (HFM)
The Thermtest Heat Flow Meter (HFM) provides users with the ability to measure thermal properties of larger, flat slab specimens ranging in thermal conductivities from 0.005 to 0.5 W/m∙K. The Heat Flow Meter is capable of measuring the steady-state heat transfer properties of foams, solids, and textiles.
Renewable sources of energy from natural resources such as solar and wind are indispensable in the fight against climate change. However, the problem associated with renewable energy is its intermittent supply. E.g., the sun doesn't shine at night and the wind doesn't...
Introduction In 1990, Maurice Ward debuted his miraculous invention on the British TV show ‘Tomorrow’s World’ by demonstrating how an easily applied paint could protect an egg from cooking after being heated by a blow torch. To the amazement of all viewers, the egg...
The Building Envelope Insulative paint can be applied to the interior and the exterior walls of a building to increase the thermal insulation. Any structure that separates the outside and the inside of a building is called the building envelope which includes...